M115 PANORAMIC TELESCOPE FINAL INSPECTION INSTRUCTIONS (CONT).
DEFLECTION ERRORS (800 MIL STEPS) AND BACKLASH (DEFLECTION MECHANISM) INSPECTIONS
Rotate deflection knob (1) until head assembly
(2) makes 2 full revolutions.
Deflection error and backlash
combined must not exceed 1.5 mil
between positive and negative error.
Total error, excluding backlash,
between deflection counter (3) and
reset counter (4) reading, for any
one number, must not exceed 0.5
Backlash in deflection mechanism
must not exceed 1.0 mil when read
on azimuth counter (5) and 1.25 mils
on reset counter (4).