ARMY TM 9-1240-379-34
MARINE CORPS TM 04332A-34
Damage may result if image in-
tensifier is exposed to bright
sunlight or high intensity ar-
Install image intensifier (37) in elbow
Install new preformed packing (38) into
threaded adapter (39).
Install threaded adapter (39) on elbow
assembly (9) and tighten. Secure by
tightening three setscrews (40).
Install eyepiece assembly (41) on
threaded adapter (39). Tighten eyepiece
assembly (41) and secure with two
Install optical eyeshield (43) onto
eyepiece assembly (41).
After maintenance has been completed,
purge and charge tank periscope in ac-
cordance with TM 750-116.
2-8. PRISM CELLMAINTENANCE INSTRUCTIONS.
THIS TASK COVERS:
b . Inspection/cleaning/repair
Artist's brush (item 3, app B)
M32E1 and M32CE1 tank periscopes
Tools and Special Tools
Helicoil insert and tool kit (4131-04-1)
T M 9-1240-379-34P
T M 750-116
Instrument and fire control shop equip-
ment (SC 4931-95-CL-A07)
Instrument and fire control system repair
shop equipment (SC 4931-95-CL-A09)